- Nov 21, 2018 -
The five indicators of the ice machine monitor the following conditions:
The green light is on: the machine is powered on;
Yellow light is on: the refrigerator is full of ice, and the machine automatically resumes work 10 seconds after the ice is taken away;
Yellow light: The water tank is short of water. When the water comes in 10 seconds, the machine automatically resumes work;
Red light is on: the condenser temperature is too high or the ambient temperature is too low;
Red light flashes: 3 minutes delay status at boot (normal);
Red light is on: the ice drill is turning wrong or the speed is lower than 850.r.P.m;
The red light is shining: the evaporator temperature is high: the evaporation temperature has not dropped below -1 °C after 10 minutes of starting;
1. The reason why the condenser temperature is high: the fan does not work; the condenser is blocked; the condensing temperature sensor or the P.C plate is damaged and the Z ambient temperature is too high.
2. Reasons for high evaporation temperature: refrigerant shortage; condensation temperature is too high; evaporation temperature sensor or P.C board is damaged.
Liquid Nitrogen Container Main Feat...
May 04, 2017
Tianchi Ice Makers Manufacturers
May 05, 2017
In Operation General Security Infor...
May 06, 2017
Liquid Nitrogen Use Matters Needing...
May 08, 2017
TianChi Ice Machine Operation Proce...
May 09, 2017
Two Methods To Check Pregnant Ewe W...
May 10, 2017
The Use Of Ice Machine Matters Need...
May 11, 2017
Liquid Nitrogen Dangerous
May 15, 2017
General Tips For Animal Ultrasound ...
May 16, 2017
Liquid Nitrogen Safety Operation Me...
May 18, 2017
The Problems And Deficiencies Of Us...
Nov 23, 2018
Cryopreservation Of Bovine Frozen S...
Nov 26, 2018
Precautions Before Using The Ice Ma...
Nov 28, 2018
Advantages Of Veterinary B-ultrasou...
Nov 30, 2018
Bovine Semen Transport And Preserva...
Dec 03, 2018
Ice Machine Cleaning And Maintenance
Dec 05, 2018
Time Problems And Re-examination In...
Dec 07, 2018
Ice Machine Mobile Precautions
Dec 12, 2018
Instrument For Measuring Pig Backfa...
Dec 14, 2018